|Minimum Order Quantity||1 Piece|
|Application||ICU , Surgical , Cardiovascular|
|Types||Automatic External Defibrillators |
Model HeartStart FRx DefibrillatorModel Number 861304How Supplied Defibrillator,battery (1),SMART Pads II (1 set),Owner Manual,Quick Reference GuideWaveform Truncated Exponential Biphasic. Waveform parametersadjusted as a function of each patient’s impedance
Energy Single energy output:
Adult:nominal 150 Joules into a 50-ohm loadInfant/Child:nominal 50 Joules into a 50-ohm loadProtocol Device follows preconfigured settings.Defibrillation andCPR protocol can be customized using HeartStart EventReview or HeartStart Configure Software
User interfaceInstructions Detailed voice prompts and visual icons guideresponder through use of the defibrillatorCPR Coaching Voice coaching for adult and infant/child CPR providesinstructions and audio cues for each breath,as well asthe appropriate number,rate and depth of chestcompressions.
Controls Green on/off button,blue i-button,orange shock
button,optional infant/child key
Indicators Ready light,blue i-button,caution light,illuminated pads,icons,shock button lights up when shock is advised.
PhysicalSize 2.4 x 7.1 x 8.9 inches (6 x 18 x 22 cm) H x D x WWeight With battery and pads case:3.5 lbs.(1.5 kg)Without battery or pads case:2.6 lbs.(1.2 kg)
EnvironmentalSealing Waterjet proof IPX5 per IEC60529Dust protected IP5X per IEC60529Temperature Operating/Standby:32° - 122° F (0° - 50° C)Altitude 0 to 15,000 feetAircraft Device:RTCA/DO-160D;1997Crush 500 poundsVibration Operating:meets MILSTD 810F Fig.514.5C-17,random;Standby:meets MILSTD 810F Fig.514.5C-18,swept sineEMI (Radiated/Immunity) CISPR II Group I Class B,IEC 61000-4-3,
and IEC 61000-4-8
Patient analysis systemPatient Analysis Evaluates patient ECG to determine if a rhythm isshockable.Rhythms considered shockable are VentricularFibrillation (VF), and certain ventricular tachycardias (VT)associated with lack of circulation.For safety reasons,some VT rhythms associated with circulation will not beinterpreted as shockable,and some very low-amplitude orlow-frequency rhythms will not be interpreted asshockable VF.
Sensitivity/Specificity Meets AAMI DF80 guidelines and AHArecommendations for adult defibrillation (Circulation1997;95:1677-1682)
Shock Advised Able to deliver a shock as soon as the device indicates
a shock is advised
Quick Shock Able to deliver a shock after the CPR pause,typically
in eight seconds
- Pay Mode Terms: T/T (Bank Transfer),Other
- Port of Dispatch: Delhi
- Production Capacity: 100
- Delivery Time: 30 days
- Packaging Details: Normal packing